Chromatin Immunoprecipitation: Chromata ChIP Histone H3 [Monomethyl Lys9] ChIP Kit [NBP1-71713] - 2 ug of NB21-1023 was used to IP DNA from fixed HeLa cells along side a no antibody (No Ab) control. DNA was measured by ...read more
Chromatin Immunoprecipitation: Chromata ChIP Histone H3 [Monomethyl Lys9] ChIP Kit [NBP1-71713] - Chromata ChIP Histone H3 [Monomethyl Lys9] Kit [NBP1-71713] - qRT-PCR standard curve generated from a serial dilution of ...read more
Chromatin Immunoprecipitation: Chromata ChIP Histone H3 [Monomethyl Lys9] Kit [NBP1-71713] - PCR product melt curve of 1% chromatin immunoprecipitation input sample run in duplicate.
This ChIP kit is specific for Histone H3 monomethyl Lysine 9, and contains all the necessary components to perform 25 Immunoprecipitation experiments and a positive control antibody (NB21-1023SS). This kit comes with a full size vial of Histone H3 Monomethyl Lys9 Antibody (NB21-1071).
This product is for research use only and is not approved for use in humans or in clinical diagnosis. Kits are guaranteed for 6 months from date of receipt.
Publications for Chromata ChIP Histone H3 Kit (NBP1-71713)(2)
We have publications tested in 1 application: ChIP.
Filter By Application
Filter By Species
Showing Publications 1 - 2 of 2.
Publications using NBP1-71713
Fishman J, Homon A, Lamsa E et al. The Complete Chemical Synthesis of Histones H3 and H4 Containing Epigenetic Modifications and Their Use in Characterizing Arginine Methylated Histone Antibodies. Poster presented at ASCB. 21st Century Biochemicals, Marlborough, MA 01752, Novus Biologicals, LLC, 8100 Southpark Way, Littleton, CO 80120.
FAQs for Chromata ChIP Histone H3 Kit (NBP1-71713). (Showing 1 - 1 of 1 FAQs).
I have one question about ChIP. In your Chromata ChIP kit, you add 25ul magnetic beads to the samples. Is there any way to normalize for differences in the amounts of magnetic beads in each sample? For example, it can happen that one sample got a bit more beads than another, is there some way to take this into account when you analyze your data?
As long as the vial/bottle of beads is vortexed so that the beads are fully resuspended in solution before taking the 25ul out, there will not be any significant variation in bead number. However, if you are taking multiple 25ul aliquots out, it is best to quickly revortex before every aliquot pipetting. For all of our internal testing, we typically see a CV of less than 2% for both intra- and interassay variation, which is very good.